AGILENT PNA-X LABVIEW DRIVER DOWNLOAD
In the Read Data. I now know enough to know that I have no clue about anything at all. These files are saved in the local hard disk of the unit and are loaded when needed. Accepted by topic author high-Q. When you get an error you should also include it’s number and details in your post. What do you mean by opening up the programmer’s guide?
|Date Added:||8 February 2012|
|File Size:||51.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 3 of If you don’t read that last byte, the next VISA read operation will get the that new-line as the first character, assume it is the Termination Character, and you get and empty string back. It looks like you wire the VISA session in as an initial string in your “format into string” block. So one of the problems is I need it to be an array of all the frequency values instead of pa-x string.
In the Read Data. I have been programming for the ENA labviee some time. Message 8 of I use a vector network analyzer.
I have gone over the web www. I am currently using HPVee 7. I am trying to establish a connection to it from my laptop. If this is the Sub VI that is throwing the error, it would be nice to know what step in the execution is failing. Message 10 of Accelerate your time to market and lower your business risks Focus on your core competencies instead of testing and data acquisition needs Utilize our certified LabVIEW developers and access our expertise and experience with up-to-date test and measurement technologies Communicate with our single point of contact for the duration of the program Improve performance with our systems, onsite installation and support Achieve the flexibility to scale systems up or down to meet demand Reduce the cost of overhead, manufacturing and logistic services Trust our 20 labiew delivery record of over 1, products and systems Leverage our Lbaview Instruments technology partnership.
Message 4 of The actual amount of bytes to read is embedded in the header.
Pna can´t read port3 and port4 – Discussion Forums – National Instruments
My support system ensures that I don’t look totally incompetent. Message 5 of He said an Agilent AE set it up and it wo Did you set the channel to 1? This is just a lazy request: Somebody asked a question regarding this last year, but I couldn’t find any follow-up. I have downloaded the driver for my instrument and even ran some of the sample programs.
Obtaining Frequency data from Agilent Analyzer – Discussion Forums – National Instruments
I will get you an error code today for it. What is the equivalent to setting all the calibration parameters on that. For any given measurement, port 1 is always the source but I may have the retu This causes problems because the read data. Message 2 of I am trying to use the following code to get data from an ENA.
Use that as the input to the VISA read byte count. I have found that if you run the following command is test.
PNA-X N5244A Labview Driver
Over the years, our team gained a valuable knowledge in implementing engineering requirements including applications, drivers, calibration procedure, self-test and verification, networking i. Message 9 of Get a feed of this content Use this view in a tile.
Type to filter by text Filter by tag Sort Sort by date created: Message 6 of On this vi the default channel value is 0, if agillent don’t change it it will probably trig an error.